H52PT-6009 USB DRIVER DOWNLOAD
You may deselect Auto-scale chart in the window. In this dialog window, select the option for automatic install. Input high is between 2. Withoutabox Submit to Film Festivals. Note that information about an item in the window is shown in the sub-window down right see Figure
|Date Added:||24 September 2011|
|File Size:||65.14 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
However, here I assume that you select Cancel.
To see the details about a virtual channel, you can click the Show Details buttoncf. The New Data Acquisition Device dialog window. Tasks and virtual channels can be given any name e. Have one to sell? There was a problem filtering reviews right now. To download, select the best match for your device and then click the Download button.
Warranty & Support
If a global channel is reconfigured, all tasks which involve that global channel are updated accordingly. On-board voltage sources available at individual terminals: We will perform a very simple test, which h52p-t6009 sufficient if we are to check that both AO0 and AI0 work correctly. Figure 20 shows the contents of the DAQmx palette.
To prepare for the uusb test, we wire together AI0 and AO0. The following relevant documents can be found at http: The video replaces the present document. Sorry, we’re having trouble showing recommendations right now. This shows the physical channel associated with the virtual channel, see Figure After the testing has passed successfully cf.
To prepare for the loopback test, AI0 and AO0 are wired together.
National Instruments NI USB Low-cost Multifunction DAQ | eBay
Download Alexa for your Windows 10 PC for free. In Figure 11, an output voltage of 3. English Choose a language for shopping.
Receiving a certification tends to be more a it takes him forever to get to the. Selecting Create Task in the menu opens a window where you can select the measurement type, see Figure The field attached to the bottom of the function is the Polymorphic VI Selector from which you can select the type of data the function will read. There was a problem completing your request.
I found out it was a Puretek PT and downloaded this driver from their website. Hope this works for you as well “.
You can use the Index Array function on the Array palette. View Cart Proceed to checkout.
In the following only the name USB is used. Arduino Leonardo with Headers.
H52pt usb driver Top Freeware Programs
Top Reviews Most recent Top Reviews. The Analog Input dialog window in the Test Panels window. The test procedure, which is denoted loopbackis to connect the AI0 channel to the AO0 channel. There ysb be a small difference between the values due to the limited resolution in the DA-converter digital-to-analog and in the AD-converter analog-to-digital. Window where you select the measurement type. You may therefore select one of the upper two options.